1. Start the foot switch before installing the mobile phone, spray out the residual air and water in the pipe, and observe whether the outlet water is normal;
2. Install mobile phones and needles;
A. Lock needle type (screw type): put the mobile phone on the treatment table, install the car needle, and screw it gently with the mobile phone key. Do not use too much force, and it can be in place.
B. Push-button type (gland type): press the button on the head of the mobile phone to load and unload the needle. Release the button and the needle will be clamped. After installing the needle, manually test whether it is installed in place.
3. Start the foot switch, adjust the water to the appropriate flow, and adjust the air to the appropriate pressure. The standard pressure of high-speed mobile phone is 0.2-0.23Mpa.
Note: Do not step on the foot switch before the mobile phone is loaded with needles, otherwise the clamping mechanism will be easily damaged.
1. When using the mobile phone, you must be careful, careful and slight. It is strictly forbidden to operate the mobile phone in a barbaric manner. Do not drop the mobile phone to the ground;
2. Regardless of the angle, it should follow the way of lifting and pulling up and down, so as to touch gently, jog, and do not force laterally.
3. The improved pen-holding mobile phone is used to find a stable fulcrum and operate intermittently.
4. According to the different tooth position and the different shape of the spare hole, replace the mobile phone and the needle in time to minimize the wear of the mobile phone bearing.
5. To use sharp needles, blunt needles not only take time and effort, but also cause damage to mobile phones and increase the pain of patients.
6. In the process of using the mobile phone, you should always check whether the needle is loose to avoid flying needles.
7. Choose different mobile phones according to different clinical situations (such as mini mobile phones that do not affect the vision of the posterior teeth for pulp opening; large torque mobile phones with strong force are used in dental preparation)
8. The one-time workload is too large, so two mobile phones should be used in turn to avoid damaging the mobile phone due to excessive load, and to save the time to replace the mobile phone.
9. The mobile phone movement has serious radial swing, and it is forbidden to use it when it moves longitudinally. It is forbidden to use the mobile phone when there is no water or water shortage.
10. The air pressure and water pressure will change with the use of the treatment table. Therefore, the air pressure and water pressure should be checked frequently during clinical operation to ensure the normal use of the mobile phone.
The mobile phone should be cleaned as soon as possible after use, otherwise the dirt will solidify and it is difficult to clean. Before unloading the needle, clean the debris on the working head with water mist, and then scrub with 75% alcohol. Or use a special automatic cleaning machine for cleaning. The method used depends on the specific situation of each clinic.
After use, just loosen (turn) the key by 1/4 turn when removing the needle. If the key is loosened too much, the three-leaf spring will push against the rear cover and be damaged. After each use, take out the needle in time to facilitate the recovery of the mechanical part of the mobile phone clip.
Maintenance (oil filling)
The pneumatic injection tank is the most ideal lubricating tool. It can not only lubricate the mobile phone, but also achieve the effect of cleaning the bearing and wind wheel.
1. Before using the fuel injection tank, first remove the needle (the spiral mobile phone can be turned back by 1/4 turn), wrap the machine head with paper, and insert the nozzle of the fuel tank into the air inlet of the mobile phone to ensure sufficient air pressure.
2. The pneumatic injection tank must be used vertically for 2-3 seconds each time.
3. 2-hole mobile phone, the nozzle is aligned with the large hole; The 4-hole mobile phone nozzle aims at the second hole.
4. In the case of oil dripping method, drop 4 drops of oil into the large hole, and then use an air gun to blow air intermittently into the oil filling hole for 30 seconds.
5. It is forbidden to use industrial lubricating oil for lubrication to prevent poisoning of patients.
6. Mobile phones that are not used for a long time should be fully oiled, sealed and stored.
Low-speed mobile phone
Whether the machine is straight or bent, it is forbidden to test the mobile phone without the needle. The machine can be tested only after the needle is installed and the fork is fastened tightly.
The fueling method of low-speed mobile phone should be added to the front half of the mobile phone. After the fueling is completed and aerated, it can be used with straight and curved handpieces. (The disinfection of mobile phones will not be described in detail here) There are certain operating procedures and use methods for any tool. Only by using it in the correct way can the performance and life of the tool be guaranteed, and the same is true for mobile phones.
I hope you can use your mobile phone according to its correct use method, and sum up your own better use method according to your own actual operation experience. In this way, your mobile phone will be better used, and its life will be longer.