Improper maintenance of dental treatment chairs not only brings a bad medical experience to patients, reduces the impression of hospitals and clinics, but also threatens the physical health of medical staff. The daily maintenance of chairs is important in steps, but it is important to persist.
1. Start cleaning: Flush the instrument hoses and the inside of the water intake system for 2 minutes.
Flush the suction hose and the interior of the suction system, with each hose sucking a cup of clean water.
Purpose: To ensure that the dental chair is in optimal working condition and provide strong support for treatment.
2. Cleaning after treatment: Flush the instrument hoses and the interior of the water intake system, and rinse for 20 seconds at all extraction points.
Flush the suction hose and the interior of the suction system, suck a cup of clean water from each hose, and disinfect if necessary.
Manual cleaning and wiping disinfection:
-External suction hose and external instrument hose - spittoons and screen covers - equipment surfaces, mats, and lighting
The handle of the doctor unit and the detachable handle of the oral lamp and oral lamp. Used for thermal disinfection or sterilization.
Purpose: After treatment, clean and maintain in a timely manner to prevent residual bacteria from spreading to patients and medical staff through air, water, and touch channels.
3. Clean every day.
Flush the instrument hoses and the inside of the water intake system, and rinse for 20 seconds at all extraction points.
Check the filter screen and replace it promptly if necessary.
Clean and disinfect the suction tube, suction system, and inside of the spittoon.
Wipe and disinfect the exterior of the straw and phone duct.
Dry the hose joint, remove the sleeve and perform thermal disinfection, wipe and disinfect.
Purpose: Thoroughly clean and check the condition of equipment every day to ensure a safe working environment.
4. Long term maintenance, key maintenance
Manual deep sterilization:
Inside the instrument hose
Water diversion system
External suction pipe
Purpose: Reasonable regular maintenance can ensure that the dental chair is as long as the new one.
Check and clean the filter bracket or replace the filter.
Remove the silicone pad for thermal disinfection.
Manually wipe and clean the cup holder.
Manually wipe and disinfect the phone holder.
Purpose: Accessories have a service life cycle. If problems are found and replaced in a timely manner, it can effectively maintain the best working condition of the equipment.
Pay attention to disinfection and stay away from infection: Surface disinfectants can quickly kill 18 pathogenic microorganisms, including Mycobacterium tuberculosis, multidrug-resistant bacteria, avian influenza, hepatitis B, hepatitis C, and HIV.
To achieve simultaneous removal of dirt and surface disinfection, without any corrosion to the surface of the object, no stimulation to the human body, and no sensitization effect.
3、 Maintenance of Air Compressor for Dental Treatment Chair
1. Power outage
Remember to cut off the power supply after work every day to prevent it from continuing to work.
2. Waste gas
After daily work, when the air compressor runs out, turn off the main valve switch! It has always been easy to accelerate the aging of the trachea, weaken the elasticity of the pressure switch spring, and directly affect the working air pressure.
3. Sewage discharge
Pay attention to sewage discharge. The air contains moisture. According to the local air humidity, it should be discharged about once a month (once every six months when the workload is heavy).